Heart Rate Measurement Using Garmin & Polar Wearables
A study was made of the Garmin Vivoactive HR and Polar H10 chest strap in terms of comparative heart rate assessments. Three different types of tests were conducted while the author wore these devices. The units are shown in Figure 1. The Garmin unit is able to be used with a number of sports, including rowing, and provides measurements of heart rate, stroke rate, distance per stroke, split times, and also provides for location tracking during the workout. Data can be uploaded to http://connect.garmin.com/and are also available for download in TCX (an XML format) as well as splits downloads in CSV format. The Polar H10 is strapped around the chest just below the level of the breast bone. This unit, too, can upload data to the http://flow.polar.com/web site, where data can be downloaded in TCX format, as well. In order to provide some variety, I considered three different activities:
- General workout, involving weight lifting, sit-ups, squats;
- Walking for 1 mile; and,
- Indoor rowing for 15 minutes.
In all cases, both the Vivoactive HR and the H10 were attached, with the Vivoactive HR snuggly affixed to the left wrist. Both watch and chest strap were properly attached with no movement between these devices and the skin. Data were collected and then downloaded and processed through a Microsoft Excel spreadsheet. The data were time-synchronized so that corresponding data points from each device were associated in time. Plots of the measurements were made.
Time-Based Plots of Heart Rate
Overlay scatter plots of heart rate measurements versus time were made of all three activities, shown in Figure 2 through Figure 4. Data were downloaded from the Garmin & Polar cloud sites and were uploaded into MS Excel. The data were then time synchronized using visual basic to align the measurements.
Heart Rate Comparison: Walking
Measurements of heart rate were taken during a one mile walk. The heart rates were plotted against one another and the correlation coefficient was computed between the two sets of measurements. In the case of the comparison shown in Figure 5, the correlation among measurements was rather poor: the correlation coefficient was determined to be -0.54. Perfect correlation is given by the diagonal line in the figure. Interesting to note is that the data points taken from the Vivoactive HR time variance. In the case of the Vivoactive HR, in some instances, the time between measurements was as high as 47 seconds with 62 measurements in the 12-14 second interval range, whereas in the case of the Polar H10, all measurements were 1 second interval. Thus, the number (quantity) of measurements taken by the Polar H10 were far denser than those of the Vivoactive HR.
Heart Rate Comparison: General Activity
In the case of general activity, which included some weight lifting, sit-ups, leg raises and standing exercises, the heart rate comparison is as shown in Figure 6. The correlation coefficient among these measurements is a bit higher at 0.60. The variation in measurement collection time associated with the Garmin HR was even higher here, with one measurement interval as high as 88 seconds!
I have hypothesized that the wide variation in data collection time may be due to arm motion that is not experienced to the degree in walking. I also have hypothesized that the improved correlation may be due to the higher heart rate, which is more easily detected by the Vivoactive HR. We will see some supporting evidence of this in the final section on indoor rowing.
Heart Rate Comparison: Indoor Rowing
Rowing on the Concept 2 PM5 unit while wearing both the Vivoactive HR and the Polar H10 produced the results as illustrated in Figure 7. The correlation between the Vivoactive HR and the Polar H10 is much higher here, with a correlation coefficient of 0.95. Several items of note: the variation in measurements with the Vivoactive HR is much lower, with only two measurements 19 seconds apart and most measurements having 1-2 second intervals. This complies much more closely with the 1-second measurement intervals of the Polar H10. Furthermore, heart rate measurements are much higher here: some measurements as high as 165 beats/minute (during sprints). In general, corroboration between the two units is better as heart rate measurement is higher. This could be due to more accurate peripheral measurement.
Based on the limited sampling and workouts thus far, the general conclusion regarding heart rate measurement “trust” is that the Polar H10 is more reliable based on several observations: (1) data collection time variation remains consistent at 1 second; and, (2) data density remains high with no dropouts in any of the workouts. This is not a surprise in general as the conventional wisdom is that chest straps are much more reliable. Yet, I wanted to quantify this reliability using some objective measures. It should be noted that while heart rate remains somewhat questionable with the Vivoactive HR, I have found that stroke rate measurement in comparison with the Concept 2 PM5 measurement is dead on accurate (at least based on the data I have observed).